Using DiscoveryQuant 2.1.3 Quick Tune will set the DP value to 0 and then Fine Tune will fail as it has an invalid DP value. The temporary Q3 MI scan method for the quicktune will show the MH ion with 2 DP scans both at 200v though this voltage may vary.
If DiscoveryQuant™ software fails to launch, even after the software is uninstalled and reinstalled, check to see if Analyst® software is also installed. An installed version of Analyst software is a prerequisite to installing DiscoveryQuant software; DiscoveryQuant software will not launch without Analyst software.
Despite being properly registered, DiscoveryQuant™ 3.0.1 software with hotfix 2 gives the error message that it is not registered and must quit. The root cause of this error is the presence of multiple instances of DiscoveryQuant software installed on the same PC.
Compound parameters could not be uploaded in Analyze mode in DiscoveryQuant™ 3.0.1 software after switching from a local database to SQL database due to issues with the database selected as the review database.
Please login to start pinning items as your favorite content.