Both Analyst® TF Software 1.8.x and the software for the Digital PicoView® Nanospray Ion Source will work together with the Windows 10 operating system. This was tested and confirmed separately by both manufacturers, SCIEX and New Objective.
When setting up ProteinPilot™ Software or the imaging acquisition software for the MALDI TOF/TOF™ Systems, a RDA (Remote Data Access) software error message is received: "TOF/TOF Imaging Acquisition Software. The client components used to communicate with the instrument are newer than those found on the instrument. Please update the Remote Data Access service on the instrument."
A mismatch of index can occur when internal standard values from a SCIEX OS software results table are processed within a query, causing the text for the reported values to appear bolded and in red, even when there is no reason for flagging these values in the report itself. This phenomena is observed for results tables created in versions of SCIEX OS Software 1.6.1 or lower and has been fixed in versions of SCIEX OS software 1.7 or greater.
This newest version of ProteoWizard software, version 3.0.20258-97f41f698, can now convert files to the txt format, as well as the three original file types (MGF, mzML and mzXML). For SCIEX software users, the txt file might be a useful option; however, the quality of the output file from ProteoWizard has not tested.
During a search with ProteinPilot™ Software 5.0, 5.0.1 and 5.0.3, the error "Unhandled Exception: Out Of Memory Exception" appears. This issue is due to the LC/MS reconstruct crashing within the ProteinPilot Software when data sets with a high MS1 scan speed are being processed.
No, the E Lens™ Technology integrated into the SCIEX Triple Quad™ 7500 LC-MS/MS System has been designed for electrospray ionization (ESI) mode of analysis and is not compatible with atmospheric pressure chemical ionization (APCI).