For research use only. Not for use in diagnostic procedures Answer
There are two patches available one for June 2010 hotfixes and one without June 2010 hotfixes installed. Both patches correct an issue found in the Analyst® 1.5.1 software that prevents users from using the Quantitation option available within the Batch Editor when the selected quantitation method contains more than 150 transitions and control of any LC device is established through AAO (Analyst® Access Objects). In this scenario, attempting to acquire the batch results in failed acquisitions. Please submit request through http://sciex.com/request-support to request this patch.