The Power of Precision
ZenoTOF 7600 system
SCIEX 7500 system
SCIEX Triple Quad 6500+ system
SCIEX 5500+ system
QTRAP 6500+ system
QTRAP 4500 system
View all mass
Echo MS system
2.0 High Throughput Multiplexing system
View all integrated
BioPhase 8800 system
8000 Plus system
800 Plus system
Advance your research with front-end
instruments designed to help you realize the full power of your mass
spectrometer. SCIEX has the broadest portfolio of ESI-MS front-ends
can facilitate various flow rates, sample requirements and
sensitivities. No one else offers the entire range
flow, microflow, nanoflow LC-MS and even ultra-low flow
ExionLC 2.0 series
View all front-end HPLC MS
Differential mobility spectrometry (DMS) and
ion mobility spectrometry are analytical techniques used to separate
ions based on their gas phase mobility. Multiple types of ion
devices exist, such as drift tubes, traveling wave, and high-field
asymmetric waveform devices. Learn how you can separate yourself
Differential Ion Mobility.
The latest ion sources from SCIEX enable
enhanced sensitivity and robustness with greater desolvation range
across all MS platforms, from Triple Quad to QTRAP and QTOF.
Turbo V ion
OptiFlow Turbo V ion
View all ion sources
vMethod applications are pre-configured and verified LC-MS/MS
that reduce the need for method development – significantly cutting
time, effort and money to deploy a new assay. Every vMethod provides
method conditions, recommended sample prep, LC and MS conditions,
details for applicable MS/MS library databases for key applications.
and personal care products (PPCPs)
and protein bioanalysis
Explore vMethod applications
Free software trials
feature request portal
Analyst TF software
Biologics Explorer software
Molecule Profiler software
High resolution and QTRAP
libraries can dramatically enhance the quality of your analysis,
you much improved confidence in your data. With a comprehensive
at your fingertips, you can easily create methods and process
and non-targeted screening data on your complex samples, faster and
easier than ever before.
All in one
HR-MS/MS library with NIST 2017
Explore the library
Boost the performance of your mass
spectrometer and improve sensitivity, productivity, and data
iChemistry Solutions are the world's only reagents and
that are custom designed with your success in mind.
RNA 9000 Purity & Integrity kit
kit for amino acid analysis of hydrolysates
kit for amino acid analysis of physiological fluids
protein digestion solution
Protein CE-SDS Purity Analysis kit
protein assay - human induction kit
CZE rapid charge variant analysis kit
BioPhase Fast Glycan Labeling and Analysis kit
standards kit for pesticide analysis
auEChERS extraction and dispersive kit
Ampliflex keto reagent
Cleavable ICAT reagent
View all consumables
QTOF – Quadrupole
Time of Flight
QTRAP® – Triple Quad Linear
SWATH® – Data Independent
SelexION® – Differential
MicroLC – Microflow
View All Technology
From substantiating new discoveries through
to end-to-end validated workflows for high-throughput biomarker
in the clinic, thousands of scientists worldwide depend upon mass
innovation from SCIEX to advance their work in the fields of
research, omics and diagnostics.
Clinical research (coming soon!)
Clinical diagnostics (coming soon!)
View All Clinical
Pesticides & herbicides
Soil and biota
View All Environmental
How do you protect your reputation and meet today’s global food
standards? Whether you are a commercial lab or a food manufacturer,
quality of your food testing data is vital to your business. SCIEX
solutions help you meet maximum residue limits (MRLs) with
data that you can genuinely count upon. With a portfolio of
applications, your lab can quickly and easily react to diverse
Authenticity & Profiling Analysis
& Food Contact Substance Analysis
Food and Beverage Testing
How do you ensure the integrity of your
results in an industry that is never constant? By accurately
even the smallest compound angles you can deliver evidence that
SCIEX forensic analysis solutions deliver fast, highly accurate
data across a multitude of compounds and biomarkers, from the known
the new and novel.
Forensic toxicology (coming soon!)
Homeland security (coming soon!)
Cannabis and hemp potency testing
Doping control (coming soon!)
View All Forensics
Metabolic flux analysis
and SNP Analysis
View All Life
Transform the capacity and capability of your biologics pipeline
complete end-to-end solutions that make your lab more productive,
more successful. With a longstanding track record in pharma
development and manufacturing, our unparalleled application
with best-in-class hardware, software and support all integrate to
revolutionize your lab.
Native Mode Analysis
Nucleic Acid and Plasmid Analysis
View All Pharma
Clinical diagnostics (coming soon!)
ClearCore MD software
Mass Spec Operators
View All Diagnostics
The SCIEX Now Learning Hub offers the most diverse and flexible learning options available, with best-inclass content that helps you to get the most out of your instrument and take your lab to the next level. Available personalized learning paths based on the latest memory science ensure better knowledge retention, and automated onboarding and enrollment means you’ll get up and running faster.
SCIEX Now Learning Hub offers the most diverse and
flexible learning options available, with best-in-class content that
helps you to get the most out of your instrument and take your lab
the next level. Available personalized learning paths based on the
latest memory science ensure better knowledge retention, and
onboarding and enrollment means you’ll get up and running faster.
SCIEX Learning Manager provides you with the infrastructure to assign, monitor and report on your staff's
competency through a single digital platform. Effectively manage the training process for new hires, ensure
continuous staff development and access information with a single log-in to your SCIEX account.
You can browse, filter, or search our
extensive list of training offerings. Choose from over 100
eLearnings or search for an instructor-led course near you. Once you
select the course you want to take, you will be directed to
Learning Hub for enrollment (login required).
SCIEX Now Online is the Everything, Anytime
destination for all your SCIEX support needs. You can keep track of
activities that matter most to you and manage your lab in the most
efficient way possible. Extensive self-help resources like our deep
Knowledge Base, enable you to solve many problems on your own. SCIEX
is available 24/7 and your new SCIEX instruments are automatically
to your profile when you purchase.
SCIEX Now Learning Hub
Manage My Instruments
SCIEX Now New Feature Request
Software New Feature Request
Online ordering solutions
Don't have an
account? Create One
Visit your SCIEX
No one understands your needs and can support
your systems better than we can. Our mission is to help you be
successful, whether it's to repair your instrument, assist with your
workflows, or help you maximize productivity in your lab. Whatever
challenge, global SCIEX Service and Support personnel are subject
experts who are focused on mass spec and capillary electrophoresis,
you'll be able to achieve your scientific goals quickly and
Plus Suite for Your New LC-MS
Instrument Service & Support
View All Support Tools
SCIEX Now Learning Hub is much more than online
courses. For the most comprehensive option, you can select Success
Programs: personalized, blended online and in-person courses. If
needs are met by a visit from one of our training experts, you can
choose multiple Onsite Training days to get your lab running. Visit
SCIEX training center in North America, Europe or Asia for intensive
classroom and laboratory training. And your online courses are
any time, from anywhere, right here.
Login to SCIEX Now Learning Hub
Success Programs at Your
Scientist Training at Your Site
Visit all Training
If you have CE, LC or mass spec questions,
then SCIEX has the answers. SCIEX support is the single destination
your system questions. We aim to fully assist you with virtually
application of our instruments, helping you to get the most out of
lab resources and assets.
View All Request Support
As a life science researcher, you need the
tools and support to help you create the scientific foundation in
pursuit of expanding the knowledge-base, whether it is understanding
fundamental biology, finding new biomarkers, discovering ways to
our quality of life, or other areas of research. We are committed to
same goals and put the very same dedication into our work to help
address your most significant research challenges.
View All Partnership
Regulatory compliance is as paramount for us
as it is for you. That’s why we have made it easy for you to freely
reference all relevant technical and product regulatory documents.
give you confidence that, with SCIEX, you will fully comply with
legislation, adhere to your laboratory protocol and meet industry
Declaration of Conformance
Safety Data Sheets
Certificates of Analysis
View All Regulatory Documents
SCIEX supplies extensive documentation to help you prepare for, use, and maintain your SCIEX hardware and software products, and we update this documentation regularly. On the Customer Documents page, you can search for and download the latest documents for your product.
Join the SCIEX community today to
interact with your peers, share and exchange ideas, develop
your knowledge, stay up-to-date with the latest products,
post insights and questions, comment on others and receive
support. This community is designed to help you, our
customers, move science forward and get the answers you
need. We’re committed to engaging with and listening to you,
to create the best customer experience possible and to
contribute to the success of your work.
Environmental / Industrial
Food and Beverage
Life Science Research
Now Feature Requests
View All Community
Our favorite papers
Meet our executives
In the news
You've got questions. We've got experts who
can help. Contact us to find out more, talk to a specialist, explore
solutions or get expert support.
Request a quote
SCIEX success network
Request hosted catalog
Global public relations
Bosnia and Herzegovina
United Arab Emirates
United Republic of Tanzania
AB Sciex is doing business as SCIEX. © 2010-2018 AB Sciex. The trademarks mentioned herein are the property
of the AB Sciex Pte. Ltd. or their respective owners. AB SCIEX™ is being used under license. Beckman
Coulter® is being used under license. Product(s) may not be available in all countries. For
information on availability, please contact your local representative. For
research use only. Not for use in diagnostic procedures.
Download tech note (PDF)
Using the SCIEX Triple Quad™ 7500 LC-MS/MS System – QTRAP® Ready
Catherine S. Lane1, Bart Van Puyvelde2, Katleen Van Uytfanghe2, Andrea-Bhangu-Uhlmann3, Maarten Dhaenens2
1SCIEX, UK, 2Ghent University, Belgium, 3PolyQuant GmbH, Germany
Targeted peptide quantification assay has been developed to investigate the potential use of nasopharyngeal swabs in a saline matrix for the quantification of SARS-CoV-2 coronavirus proteins using LC-MS/MS. The gains in sensitivity attained by moving the assay onto the next-generation SCIEX Triple Quad 7500 LC-MS/MS System and reducing matrix effects by changing swab storage medium were characterized.
Research into the epidemiology of the SARS-CoV-2 virus necessitates wide-scale sampling of large populations. The majority of assays for viral testing target detection of the SARS-CoV-2 nucleic acid (RNA), however, antigen tests targeting viral protein are generating interest globally amongst researchers in this field. The use of mass spectrometry for the detection of proteins from biological matrices is well established. However, the desire for increased sensitivity is always a goal in any quantitative assay and, thus far, in this case, mass spectrometry has struggled to match the sensitivities achieved by the more widely used reverse-transcriptase polymerase chain reaction (RT-PCR)-based tests. Here, the sensitivity of peptide detection in swab matrices is explored in order to compare with the sensitivity typical in RNA based detection strategies.
The SCIEX 7500 System shows significant gains in sensitivity relative to the previous generation of instruments because of improvements in both the generation and the sampling of ions.1,2 Furthermore, the preservation medium utilized for the collection of swab samples can have a considerable impact on the sensitivity of the LC-MS/MS assay.3 Here, building on the method developed previously4, the MRM assay is applied to the analysis of nasopharyngeal swab samples preserved in eSwab medium (saline matrix), using the SCIEX 7500 System to quantify the increased sensitivity (Figure 1).
Figure 1. Quantification of NCAP peptide. Calibration line and extracted ion chromatograms (XICs) for KQQTVTLLPAADLDDFSK from SARS-CoV-2 NCAP protein spiked into nasopharyngeal swab samples (XICs shown are for 2.5 ng and 10 ng protein in 250 µL swab samples).
The work described here was carried out as part of the Cov-MS consortium3, a community group established by Maarten Dhaenens at Ghent University, Belgium, with the aim of increasing applicability, accessibility, sensitivity and robustness of the detection of SARS-CoV-2 by electrospray mass spectrometry, thus increasing awareness of the technique as an important research tool. The consortium is made up of both academic laboratories and industrial partners. As such, the samples analyzed here were also examined by multiple other collaborators.
Sample preparation: Additional method optimization and retention time scheduling was performed using trypsin-digested spike glycoprotein (P0DTC2, SPIKE_SARS2), nucleocapsid protein (P0DTC9, NCAP_SARS2) and QconCAT standard (PolyQuant), prepared as previously described.4
Dilution series were prepared by spiking the recombinant proteins (0, 0.05, 0.1, 0.5, 1, 5, 10, 50, 100 ng) into 250 µL pooled nasopharyngeal swabs from 3 volunteers in eSwab matrix. To all samples, 120 ng QconCAT was added. Samples were vortexed with ice-cold acetone (1750 µL), then centrifuged at 16,000g for 10 mins at 0 °C. Acetone was removed, and pellets were left to dry. Dried pellets were re-suspended in 250 µL of 5% acetonitrile in 50 mM TEAB, then 50 µL of 0.1µg/µL Trypsin/LysC was added. Samples were vortexed, centrifuged and incubated for 15 minutes at 37 °C. 50 µL of each sample was removed and lyophilized. Samples were resuspended in 50 µL of 0.1% formic acid in 5% acetonitrile and 5 µL was injected per analysis. A sample containing approximately 2.5 ng recombinant protein in 250 µL eSwab was prepared by mixing equal volumes of the 5 ng and 0.05 ng samples.
Chromatography: Peptide separation was performed using an ExionLC™ system and a Phenomenex Luna Omega Polar C18 column (2.1 x 50 mm, 1.6 µm). Flow rate of 0.6 mL/min was used for fast analysis times. Gradient is described in Table 1. Column temperature was 40 °C.
Table 1. Gradient for peptide separation.
Mass spectrometry: MRM analysis of the selected peptides was performed using a SCIEX 7500 System operating in positive ionization mode. Source conditions were ISV 2000 V, GS1 40, GS2 70, TEM 450°C and CUR 40. Scheduled MRM™ Algorithm was used for optimum detection efficiency, using retention time tolerance of 15 sec and a target scan time of 0.4 sec. MRM transitions targeting QconCAT and histone peptides were assigned dwell times of 5 msec. Optimized MRM transitions and compound dependent parameters are shown in Table 2.
Data processing: MRM transition and voltage optimization was performed using Skyline software. Source optimization was performed using SCIEX OS-Q Software. Data were processed using SCIEX OS-Q Software.
In this work, the MRM assay previously optimized on the SCIEX Triple Quad 5500+ LC-MS/MS System – QTRAP Ready4, targeting peptides from SPIKE and NCAP, was transferred to the SCIEX 7500 System to determine the increased sensitivity. Using the digested recombinant proteins and Skyline software, collision energy values optimized on the SCIEX 5500+ System were re-evaluated on the SCIEX 7500 System and found to be very similar (within 4 V). Stable-isotope labeled standards are routinely used in MS-based assays to increase robustness, allow absolute quantification, and enable normalization across multiple instruments and laboratories. The Cov-MS QconCAT construct includes seventeen 15N-labeled SPIKE and NCAP peptides, separated by their native amino acids and one peptide from each of four human core histone proteins that were added to serve as markers for swab sampling quality. It was surmised that efficient swab sampling should cause superficial tissue damage, resulting in the presence of nuclear histone proteins in the swab sample. If the histone peptides are absent from a sample, this could potentially indicate poor sampling efficiency. MRM transitions were optimized for histone peptides using the QconCAT standard and Skyline software.
The final MRM assay involved two MRMs per peptide (one MRM each for native and heavy-isotope-labeled peptides) for each of ten of the best-responding SPIKE and NCAP peptides and three histone peptides (Table 2). One SPIKE peptide, FLPFQQFGR, found to respond well elsewhere, was included in the assay despite not being present in the QconCAT standard. If this peptide were to be retained for future SARS-CoV-2 antigen analyses, its incorporation into the QconCAT structure would be desirable. Retention times were scheduled, and dwell times fixed at 5 msec for heavy-isotope-labeled and histone peptides, to allow the bulk of the cycle time to be assigned to native SPIKE and NCAP peptides.
Wide-scale detection of SARS-CoV-2 would necessarily require the ability to measure very large numbers of samples per day. In the interest of throughput, the LC column and gradient used for peptide separation was shortened compared to previous work.4 A 50-mm column with 1.6-µm particle size was selected, and the total analysis time shortened to 4.5 min (from 7 min). The shortened analysis time provided good separation of target peptides (Figure 2).
Table 2. Optimized MRM transitions for quantification of key SARS-CoV-2 coronavirus proteins, SPIKE and NCAP, on the SCIEX 7500 System.
Figure 2. Chromatographic separation of 10 SARS-CoV-2 peptides (upper panel) and 3 histone peptides (lower panel) from a pooled nasopharyngeal swab sample in eSwab matrix spiked with recombinant SARS-CoV-2 proteins and QconCAT (100 ng each of SPIKE and NCAP, 120 ng QconCAT in 250 µL swab sample). Excellent separation of target peptides was maintained despite decreasing the time of the analysis from 7 min to 4.5 min.
The preservation medium used for the collection of swab samples can have considerable impact on the sensitivity of the LC-MS/MS assay.3 Previously4, samples were analyzed in UTM viral transport medium. However, UTM contains proteins, antibiotics and antimycotics5, which cause high background signal and ion suppression in the assay. ESwab is a salt-based medium without the protein and antibiotic content.6
While ion suppression would still be expected from the eSwab background, no protein-based interference would be anticipated. With this in mind, dilution series were prepared by spiking recombinant proteins and QconCAT into pooled nasopharyngeal swabs from volunteers in eSwab matrix. Samples were subjected to acetone precipitation and protein pellets were resuspended and digested with Trypsin/LysC. Samples were analyzed on the SCIEX 7500 System using the optimized MRM assay.
Sensitivity increases of 2 to 20 fold were observed for the SCIEX 7500 System analysis of peptides from SARS-CoV-2 proteins in nasopharyngeal swabs stored in eSwab, when compared with analysis of similar samples in UTM background on the SCIEX 5500+ System4 (Table 3). Lower limits of detection and quantification (LLOD and LLOQ) are outlined in Table 3. Examples of the MRM signal observed from SARS-CoV-2 peptides in eSwab matrix are shown in Figure 3. Calibration curves for all peptides are shown in Figure 4. The typical bioanalytical criteria for determination of LLOQ are accuracy of ±20%, precision of <20% at LLOQ and <15% across the calibration curve.
An example of how use of the QconCAT internal standard can increase assay robustness during sample preparation is shown in Figure 5 and Table 4.
Table 3. Lower limits of detection and quantification obtained for 10 peptides. The LLODs ranged from 0.07 to 0.44 fmol/µL in pooled nasopharyngeal swabs in eSwab matrix on the SCIEX 7500 System. Significant improvements in detection limits were observed from the previous work, by switching to using the eSwab medium and the higher sensitivity instrument.
Figure 3. Example peptide data. Signal from four SARS-CoV-2 peptides for the low end of the dilution series.
Figure 4. Calibration curves for ten SARS-CoV-2 peptides in eSwab matrix. Excellent linearity was observed for all peptides with r2 values between 0.9940 and 0.9997.
Figure 5. Use of a QconCAT internal standard for increased assay robustness during sample preparation. The upper pane shows variation in MRM peak area for 3 histone peptides in each of 10 samples, each analyzed three times (30 consecutive analyses). The lower pane shows variation in MRM peak area ratio (analyte peak area / internal standard peak area) for the same analyses. For each peptide, the use of the QconCAT stable-isotope-labeled standard substantially reduces the %CV for the 30 analyses (Table 4).
Table 4. Variation in %CV for three histone peptides in 10 samples with and without use of the QconCAT internal standard.
While the work described here is one step further towards a LC-MS/MS assay for the measurement of viral proteins in swab samples, there is still more work to be done. Sensitivities for current RT-PCR tests are measured in cycle threshold (Ct) values. While the same sample can give different Ct values on different RT-PCR assays7 a lower Ct value indicates a higher level of virus. As the amount of virus decreases, Ct values rise gradually. Because Ct is only a semi-quantitative (and variable) indicator of the amount of virus present, the conversion from Ct to amount of viral protein in a swab sample is not straightforward. Nevertheless, recently Van Puyvelde et al.3 estimated a Ct value of 20 to be approximately equivalent to 1.3 fmol NCAP in 10 µL swab sample, and Ct 25 to be equivalent to 41 amol NCAP in 10 µL swab sample. If a 32-fold concentration factor were applied at the sample preparation stage, then 1.3 fmol and 41 amol would correspond to Ct 25 and 30, respectively. These limits of detection in matrix are achievable using high-end triple quadrupole instruments8 but, for the highest possible levels of sensitivity, optimization of sample preparation will be key. Potential areas for improvement include the substitution of eSwab matrix for one with lower salt content, the use of other sample clean-up approaches and the use of sample enrichment strategies. One such strategy, SISCAPA (Stable Isotope Standards and Capture by Anti-Peptide Antibodies), has previously been used to improve the performance of mass spectrometry for measurement of pre-selected protein targets.9
Here, the targeted peptide quantification assay described previously4 has been refined and developed to investigate the potential use of nasopharyngeal swabs in a saline matrix for the quantification of SARS-CoV-2 coronavirus proteins. Further optimization included the addition of a QconCAT stable isotope labeled standard for increased robustness, swab sampling efficiency, inter-laboratory comparison and reduction in analysis time for use in high-throughput environments. The gains in sensitivity attained by moving the assay onto the SCIEX 7500 System, and by reducing matrix effects by changing swab storage medium, could be further increased by use of sample enrichment and clean-up strategies, such as SISCAPA9 or SPE, or by moving to a more MS-friendly swab storage matrix.
This assay has not been validated on real test samples and should not be used for patient testing. More work is required to translate this peptide quantification assay into a useful assay for determination of viral particle numbers.
The Destination for All Your Support Needs