NEWS: Important Safety Notice for API 4000™, 4000 QTRAP®, and API 5000™ products. Learn More >
Knowledge Base is your tool to finding answers about common questions. Browse all Knowledge Base articles below.
If a batch completes, but the LC pumps never turn off, the Max Idle Time may need to be adjusted in Queue Options.
How to configure a CTC autosampler to use custom dimensions for use with alternative plates or vial racks.
When upgrading to Sciex OS 1.2 from version 1.0, the software for Sciex OS 1.0 must be uninstalled prior to installing Sciex OS 1.2. LibraryView Framework version 1.1 must be removed as well. Certain companion software must be removed as well, and installed again after Sciex OS 1.2 is installed.
To remove an unneeded analyte's MRM transitions from a MultiQuant software method, the quantitation method must be opened and the Delete Selected Rows function must be used.
Installing Markerview 126.96.36.199. requires an activation key code, not a license file. License activation keys can be obtained by emailing email@example.com.
Batches can be rearranged in the Analyst software queue using the Move Batch option. In the Move Batch window, drag the batch to be moved to the desired location to rearrange.
Editing a query file extension requires renaming of the file name.
None of the SCIEX TOF instruments, 5600, 5600+, 6600, X500R, or X500B, can do a precursor ion IDA experiment.
Converting a non-scheduled MRM method to a scheduled MRM method requires that the retention time of each transition is updated in the Analyst software method.
When running Analyst TF 1.7.1 in sim mode on a processing workstation, an additional driver, the Chronologic driver, must be downloaded and installed.