The SCIEX clinical diagnostic portfolio is For In Vitro Diagnostic Use. Rx Only. Product(s) not available in all countries. For information on availability, please contact your local sales representative or refer to https://sciex.com/diagnostics. All other products are For Research Use Only. Not for use in Diagnostic Procedures.
Trademarks and/or registered trademarks mentioned herein, including associated logos, are the property of AB Sciex Pte. Ltd. or their respective owners in the United States and/or certain other countries.
© 2020 DH Tech. Dev. Pte. Ltd.
Knowledge Base is your tool to finding answers about common questions. Browse all Knowledge Base articles below.
The executable file for Excel – Excel.exe – is located in the installation directory for the 64-bit version of Microsoft Office 365 at C:\Program Files\Microsoft Office\root\Office16.
The files created in SCIEX OS Software are not backwards compatible with other versions of the software. An error will appear when a processing method created with a higher version of SCIEX OS Software (e.g., 2.0) is opened with a previous version of SCIEX OS Software (e.g., 1.7). To open a processing method created in the latest version of SCIEX OS Software with the older version, export the processing method as a text file.
To add the concentrations of different isomers together when processing data in SCIEX OS Software, open a processing method in the Analytics mode of the software. Select the Calculated Columns tab in the processing method and then select Add Formula to create a custom calculation for adding the isomers.
For analytes using an internal standard, the actual retention time (RT) of the internal standard peak (for the corresponding sample) is first determined. The expected RT for the analyte is determined by multiplying the specified expected RT by the ratio of the actual to the expected RT for the internal standard.
If a user starts a batch using Analyst® MD Software and then signs out of their Windows account while Analyst Software is acquiring data, the software will close and suspend the run. It is not recommended to sign out of a user's Windows account if Analyst MD Software is acquiring data.
This article discusses the three ways to open a query (a file with the extension .query), a file linked to a report template that can be used to perform more complicated calculations or sorting functions in a results table.
One common error when using the SWATH® Acquisition MicroApp 2.0.1 for processing SWATH data is to relate the XIC width setting in the processing software with the mass accuracy of the instrument, and set the XIC width to a value that is too low. XIC extraction widths should be set to 50 to 75 ppm, regardless of the mass accuracy of the acquired data.
The steps needed to connect to a new compound database library in Analyst® Software are detailed in this article.
When reporting results table values to a comma-separated values (CSV) report, for some analytes, the reported information is shifted from the correct column to one column over to the left. This column-shifting issue is observed when an analyte name contains a comma.
For libraries imported into SCIEX OS Software and used with LibraryView 1.x Software, the license file needs to be saved to the directory C:\Program Files\AB SCIEX\LibraryView\LibraryViewFramework\Server in order to unlock the library and activate it.
A new password has been sent to your email.
You have added a user.
Unfortunately the serial number you provided shows as inactive in our database. Would you like to submit the case anyway?
This serial number could not be found. Would you like to submit the case anyway?
Your password has been updated.
Your account is now created.
A verification link has been sent to your email.
A new password has been sent to your email.
Your insturment has been removed.
Your privacy settings are now updated.
Unfortunately the serial number you provided shows as inactive in our database. Please open a support case here if you require more assistance.
Software has already been registered to another instrument. Please check your AID and try again.
If you want to change the License Key to another instrument, please submit a new case.
Software has already been registered to the instrument selected.
Please check your instrument's software by going to the Instruments page and selecting the instrument in your "My Instruments" section. Here you will see a Software tab where you can view your instrument's software information.
You can also go to the Registered Software page to view software associated with your instrument(s).
If you are not finding what you are looking for, please submit a new case.