The SCIEX clinical diagnostic portfolio is For In Vitro Diagnostic Use. Rx Only. Product(s) not available in all countries. For information on availability, please contact your local sales representative or refer to https://sciex.com/diagnostics. All other products are For Research Use Only. Not for use in Diagnostic Procedures.
Trademarks and/or registered trademarks mentioned herein are the property of AB Sciex Pte. Ltd. or their respective owners in the United States and/or certain other countries.
© 2019 DH Tech. Dev. Pte. Ltd.
Knowledge Base is your tool to finding answers about common questions. Browse all Knowledge Base articles below.
After upgrading Analyst® TF software 1.8 to version 1.8.1, the Eksigent control software no longer displayed the option to chose the ekspert™ nanoLC 400 autosampler. To resolve this issue, in the Eksigent control software, the autosampler driver for the nanoLC 400 must be unregistered and then re-registered.
Selecting the question mark in the upper right hand corner of SCIEX OS software launches the SCIEX OS Help System screen. If no information appears for a selected topic, the SCIEX OS Help.chm file located in the C:\Program Files\SCIEX\SCIEX OS\Documentation folder needs to be unblocked.
If a report generated from SCIEX OS software using a template and a query contains information about internal standard samples, the internal standard information showing in the results table does not appear correctly in the report. To resolve this issue, all internal standard samples must be listed at the bottom of the SCIEX OS software processing method in the components section.
An IF statement can be added to limit the sample types that need to be output to a report from a results table created in Analyst® software, MultiQuant™ software or SCIEX OS software. Below are few examples of how to write an IF statement within the loops ForEachSample or ForEachAnalyte.
If a report template needs to be opened for editing or if a new custom template needs to be added to the template list in SCIEX OS software, then navigate to C:\ProgramData\SCIEX\Analytics\Reporter to locate or add a template.
Transferring data to a LIMS system from a results table created in SCIEX OS software does not require a template. Once the results table is saved and locked, under the Reporting menu, select Transfer results to LIMS, and two LIMS export options will appear in the LIMS transfer dialog box.
Individual users of Analyst® software can be assigned a particular role in Analyst software in the Security Configuration mode. The various roles are designed to limit a user's access to certain functions within the software depending on the role assigned. Below is a chart that shows how access to these functions decreases for the various default roles.
To add a custom column(s) to a batch in Analyst® software, right click within the batch table on the Sample tab, and select Add Custom Column.
If the dilution factor is not set correctly in an Analyst® software results table, the calculated concentration values for samples of known concentration will be incorrect.
No, the loops, ForEachSample and ForEachVisibleSample, cannot be used within the same SCIEX OS software report template. Similarly, ForEachAnalyte and ForEachVisibleAnalyte cannot be used together in the same report template.
A new password has been sent to your email.
You have added a user.
This serial number could not be found. Would you like to submit the case anyway?
Your password has been updated.
Your account is now created.
A verification link has been sent to your email.
A new password has been sent to your email.
Your insturment has been removed.
Your privacy settings are now updated.
Software has already been registered to another instrument. Please check your AID and try again.
If you want to change the License Key to another instrument, please submit a new case.
Software has already been registered to the instrument selected.
Please check your instrument's software by going to the Instruments page and selecting the instrument in your "My Instruments" section. Here you will see a Software tab where you can view your instrument's software information.
You can also go to the Registered Software page to view software associated with your instrument(s).
If you are not finding what you are looking for, please submit a new case.