The SCIEX clinical diagnostic portfolio is For In Vitro Diagnostic Use. Rx Only. Product(s) not available in all countries. For information on availability, please contact your local sales representative or refer to https://sciex.com/diagnostics. All other products are For Research Use Only. Not for use in Diagnostic Procedures.
Trademarks and/or registered trademarks mentioned herein are the property of AB Sciex Pte. Ltd. or their respective owners in the United States and/or certain other countries.
© 2020 DH Tech. Dev. Pte. Ltd.
Knowledge Base is your tool to finding answers about common questions. Browse all Knowledge Base articles below.
To routinely add the same custom columns to each newly generated MultiQuant results table, custom columns can be added to the table in the Batch Editor in Analyst® software that is created before acquiring a data set.
To include only the data associated with the quantifier ion in a report, group the quantifier and qualifier ion for a particular analyte, and then use the For Each AnalyteGroup loop around the information to be reported.
By default, the format for a numerical field in a report is taken from the results table itself if the format is not set in the report template tag. If a specific numerical format is desired for a particular field in a report, certain default formats (such as the number of decimals desired or scientific notation) can be chosen in the Format field in the tag for that value. Otherwise, In the SCIEX OS software results table itself, the number format can be set for individual columns by editing the Number Format in the Results Table Display Settings options.
After upgrading Analyst® TF software 1.8 to version 1.8.1, the Eksigent control software no longer displayed the option to chose the ekspert™ nanoLC 400 autosampler. To resolve this issue, in the Eksigent control software, the autosampler driver for the nanoLC 400 must be unregistered and then re-registered.
An IF statement can be added to limit the sample types that need to be output to a report from a results table created in Analyst® software, MultiQuant™ software or SCIEX OS software. Below are few examples of how to write an IF statement within the loops ForEachSample or ForEachAnalyte.
Individual users of Analyst® software can be assigned a particular role in Analyst software in the Security Configuration mode. The various roles are designed to limit a user's access to certain functions within the software depending on the role assigned. Below is a chart that shows how access to these functions decreases for the various default roles.
To add a custom column(s) to a batch in Analyst® software, right click within the batch table on the Sample tab, and select Add Custom Column.
If the dilution factor is not set correctly in an Analyst® software results table, the calculated concentration values for samples of known concentration will be incorrect.
If the name of a query is ever changed or updated or if a new query needs to be attached to a report template, the query must be relinked to the template used for reports created in MultiQuant™, Analyst® or SCIEX OS software.
If actual concentration values for standards are omitted from a MultiQuant™ or Analyst® software results table, then the data points for standard concentration versus peak area will all be aligned with zero on the x-axis in the calibration curve, looking nothing like the expected diagonal line. Adding the actual concentration information for the standards spreads the points across the x-axis and restores the expected diagonal linear regression line.
A new password has been sent to your email.
You have added a user.
Unfortunately the serial number you provided shows as inactive in our database. Would you like to submit the case anyway?
This serial number could not be found. Would you like to submit the case anyway?
Your password has been updated.
Your account is now created.
A verification link has been sent to your email.
A new password has been sent to your email.
Your insturment has been removed.
Your privacy settings are now updated.
Unfortunately the serial number you provided shows as inactive in our database. Please open a support case here if you require more assistance.
Software has already been registered to another instrument. Please check your AID and try again.
If you want to change the License Key to another instrument, please submit a new case.
Software has already been registered to the instrument selected.
Please check your instrument's software by going to the Instruments page and selecting the instrument in your "My Instruments" section. Here you will see a Software tab where you can view your instrument's software information.
You can also go to the Registered Software page to view software associated with your instrument(s).
If you are not finding what you are looking for, please submit a new case.