The Power of Precision
ZenoTOF 7600 system
SCIEX 7500 system
SCIEX Triple Quad 6500+ system
SCIEX 5500+ system
QTRAP 6500+ system
QTRAP 4500 system
View all mass
Intabio ZT system
Echo MS system
2.0 High Throughput Multiplexing system
View all integrated
BioPhase 8800 system
8000 Plus system
800 Plus system
Advance your research with front-end
instruments designed to help you realize the full power of your mass
spectrometer. SCIEX has the broadest portfolio of ESI-MS front-ends
can facilitate various flow rates, sample requirements and
sensitivities. No one else offers the entire range
flow, microflow, nanoflow LC-MS and even ultra-low flow
ExionLC 2.0 series
View all front-end HPLC MS
Differential mobility spectrometry (DMS) and
ion mobility spectrometry are analytical techniques used to separate
ions based on their gas phase mobility. Multiple types of ion
devices exist, such as drift tubes, traveling wave, and high-field
asymmetric waveform devices. Learn how you can separate yourself
Differential Ion Mobility.
The latest ion sources from SCIEX enable
enhanced sensitivity and robustness with greater desolvation range
across all MS platforms, from Triple Quad to QTRAP and QTOF.
Turbo V ion
OptiFlow Turbo V ion
View all ion sources
vMethod applications are pre-configured and verified LC-MS/MS
that reduce the need for method development – significantly cutting
time, effort and money to deploy a new assay. Every vMethod provides
method conditions, recommended sample prep, LC and MS conditions,
details for applicable MS/MS library databases for key applications.
and personal care products (PPCPs)
and protein bioanalysis
Explore vMethod applications
feature request portal
Biologics Explorer software
Molecule Profiler software
High resolution and QTRAP
libraries can dramatically enhance the quality of your analysis,
you much improved confidence in your data. With a comprehensive
at your fingertips, you can easily create methods and process
and non-targeted screening data on your complex samples, faster and
easier than ever before.
All in one
HR-MS/MS library with NIST 2017
Explore the library
Boost the performance of your mass
spectrometer and improve sensitivity, productivity, and data
iChemistry Solutions are the world's only reagents and
that are custom designed with your success in mind.
RNA 9000 Purity & Integrity kit
kit for amino acid analysis of hydrolysates
kit for amino acid analysis of physiological fluids
Protein CE-SDS Purity Analysis kit
MS calibration kits
CZE rapid charge variant analysis kit
BioPhase Fast Glycan Labeling and Analysis kit
View all consumables
QTOF – Quadrupole
Time of Flight
QTRAP® – Triple Quad Linear
SWATH® – Data Independent
SelexION® – Differential
MicroLC – Microflow
View All Technology
ADME and DMPK
Viral vector characterization
High-throughput mass spectrometry
Targeted protein degraders and PROTACs
Extractables and leachables
Residual nucleic acid analysis
Lipid nanoparticles and non-viral carrier
Messenger RNA analysis
Plasmid and DNA analysis
Synthetic oligonucleotide analysis
Cell line analysis
Charge heterogeneity analysis
Intact protein analysis
Peptide mapping analysis
Subunit mass analysis and middle-down
View all Biopharma and pharma
Clinical Mass Spec Operators
Clinical Method Developers
Clinical Lab Managers
View All Clinical
Pesticides & herbicides
Soil and biota
View All Environmental
How do you protect your reputation and meet today’s global food
standards? Whether you are a commercial lab or a food manufacturer,
quality of your food testing data is vital to your business. SCIEX
solutions help you meet maximum residue limits (MRLs) with
data that you can genuinely count upon. With a portfolio of
applications, your lab can quickly and easily react to diverse
Authenticity & Profiling Analysis
& Food Contact Substance Analysis
Food and Beverage Testing
How do you ensure the integrity of your
results in an industry that is never constant? By accurately
even the smallest compound angles you can deliver evidence that
SCIEX forensic analysis solutions deliver fast, highly accurate
data across a multitude of compounds and biomarkers, from the known
the new and novel.
Homeland security (coming soon!)
Cannabis and hemp potency testing
View All Forensic
Metabolic Flux Analysis
and SNP Analysis
View All Life
The SCIEX Now Learning Hub offers the most diverse and flexible learning options available, with best-inclass content that helps you to get the most out of your instrument and take your lab to the next level. Available personalized learning paths based on the latest memory science ensure better knowledge retention, and automated onboarding and enrollment means you’ll get up and running faster.
SCIEX Now Learning Hub offers the most diverse and
flexible learning options available, with best-in-class content that
helps you to get the most out of your instrument and take your lab
the next level. Available personalized learning paths based on the
latest memory science ensure better knowledge retention, and
onboarding and enrollment means you’ll get up and running faster.
SCIEX Learning Manager provides you with the infrastructure to assign, monitor and report on your staff's
competency through a single digital platform. Effectively manage the training process for new hires, ensure
continuous staff development and access information with a single log-in to your SCIEX account.
You can browse, filter, or search our
extensive list of training offerings. Choose from over 100
eLearnings or search for an instructor-led course near you. Once you
select the course you want to take, you will be directed to
Learning Hub for enrollment (login required).
Login to SCIEX Now Learning Hub
Success Programs at Your
Scientist Training at Your Site
Visit all Training
SCIEX Now Learning Hub
Manage my instruments
SCIEX Now New Feature Request
Visit your SCIEX
Declaration of conformance
Safety data sheets
Certificates of analysis
View All regulatory documents
Software and IT services
LC-MS service plans
Protect Plus Suite for your new LC-MS
CE service plans
Clinical service plans
StatusScope remote monitoring
Software accelerator program
Software support plans
Software support policy
Premium access content
Academic partnership program
View all partnership programs
Join the SCIEX community today to
interact with your peers, share and exchange ideas, develop
your knowledge, stay up-to-date with the latest products,
post insights and questions, comment on others and receive
support. This community is designed to help you, our
customers, move science forward and get the answers you
need. We’re committed to engaging with and listening to you,
to create the best customer experience possible and to
contribute to the success of your work.
Environmental / Industrial
Food and Beverage
Life Science Research
Now Feature Requests
View All Community
Echo® MS Center Of Excellence
Our favorite papers
Meet our executives
In the news
You've got questions. We've got experts who
can help. Contact us to find out more, talk to a specialist, explore
solutions or get expert support.
Request a quote
SCIEX success network
Request hosted catalog
Global public relations
Bosnia and Herzegovina
United Arab Emirates
United Republic of Tanzania
AB Sciex is doing business as SCIEX. © 2010-2018 AB Sciex. The trademarks mentioned herein are the property
of the AB Sciex Pte. Ltd. or their respective owners. AB SCIEX™ is being used under license. Beckman
Coulter® is being used under license. Product(s) may not be available in all countries. For
information on availability, please contact your local representative. For
research use only. Not for use in diagnostic procedures.
Download tech note (PDF)
Site-localization of malonylated peptides using the SCIEX ZenoTOF 7600 system
Joanna Bons1, Jason Cason2, Birgit Schilling1, Christie Hunter3
1Buck Institute, USA, 2SCIEX, Canada, 3SCIEX, USA
Determining the identity and the precise location of a post-translational modification on a protein is important to fully characterize function. Here, the utility of electron activated dissociation on the SCIEX ZenoTOF 7600 system was investigated for the characterization of labile post translational protein modifications. Leveraging the tunable kinetic energy, the malonyl modification on a synthetic peptide was preserved such that definitive localization was possible. In addition, the use of Zeno trapping provided significant sensitivity gains to ensure complete sequence characterization was obtained.
Post-translational modifications (PTMs) are important players in a diverse group of functions that include protein conformation and signaling. Lysine acylation, such as malonylation, is one such PTM, and is regulated in part by lysine deacylases, which are members of the sirtuin (SIRT) protein family. In a previous study investigating SIRT5-regulated lysine malonylome, it was shown that 183 malonyllysine sites (from 120 proteins) out of the 1,137 identified sites (from 430 proteins) were significantly increased in Sirt5-/- KO versus wild-type mice.1 Specifically, it revealed that malonylation regulated GADPH activity. Malonylated peptides are however traditionally difficult to characterize using mass spectrometry and CID because the modification is extremely labile.
In this work, that involves using EAD technology with tunable kinetic energy, the effects of kinetic energy ramping on the preservation of highly labile PTMs (malonylation, for example) was studied, focusing on one previously identified malonyllysine site from GADPH (K-192). Two orthogonal fragmentation modes were compared (EAD vs. CID), to investigate the utility of each for PTM site localization. In addition, samples were measured using MRMHR mode to investigate the use of EAD for quantitative PTM characterization of labile modifications.2 MS/MS data were acquired with the Zeno trap activated, which provides significant sensitivity increases and enhances the quality of EAD MS/MS spectra.
Figure 1. Sequence of the investigated malonylated peptide. Peptide at position 185-195 in mouse glyceraldehyde-3-phosphate dehydrogenase (P16858) and carrying a malonyl group on lysine K-192 was investigated. PTM site-specific ions are highlighted in blue.
Sample preparation: Malonylated synthetic peptides were obtained from Thermo Fisher Scientific and diluted in simple matrix for analysis.
Chromatography: A NanoLC 425 system plumbed for microflow chromatography (5 µL/min) was used and operated in direct inject mode. The analytical column used was a 0.3 mm x 150 mm (2.6 µm particle size) Phenomenex Omega Polar column. Column temperature was controlled at 30°C. Short gradients of 8 minutes were used.
Mass spectrometry: All data were acquired using a SCIEX ZenoTOF 7600 system and an OptiFlow Turbo V ion source4 equipped with the microflow probe and 25 µm electrodes. The system is equipped with the electron activated dissociation (EAD) cell which enabled electron-based fragmentation to be performed in a targeted manner on the malonyl peptides. MRMHR Acquisition data were collected using a TOF MS scan of 250 msec and MS/MS accumulation times of 150 msec both in CID and EAD mode. Kinetic energies were ramped and optimized per analyte. Electron current was also ramped to optimize and a final value of 5000 nA was used throughout study.
Data processing: The MRMHR data was processed using SCIEX OS software 2.1 using both Explorer and Analytics modules.
Malonylated peptide TVDGPSGKmaLWR from mouse glyceraldehyde-3-phosphate dehydrogenase (P16858) was synthetized with a malonylated lysine at position K-192 (Figure 1). MRMHR analysis using CID fragmentation mode caused significant neutral loss of CO2 (-44 m/z) from the labile malonyl group for the y-ion series (Figure 2A). However, EAD fragmentation generated intact z+1-ion and c-ion series, that were easily detectable in the MS/MS spectra (Figure 2B).5 Specifically, fragment ions were observed, ranging from z4+1 to z11+1, or also c9 and c10 that contained the labile PTM, that did not undergo a significant neutral loss, enabling confident PTM site localization.
Using the Peptide fragment pane in Bio Tool Kit, the fragment ion coverage is easily visualized between the two dissociation techniques (Figure 2C).
Figure 2. MS/MS spectra of TVDGPSGKmaLWR peptide analyzed in CID and EAD modes. The malonylated peptide (m/z 656.3320) was analyzed in (A) CID mode and (B) EAD mode (KE = 5 eV). CID fragmentation resulted in neutral loss of -CO2 (-44), and no “intact” PTM-specific differentiating ions were detectable. Comprehensive EAD fragmentation generated high-intensity fragment ions that provided evidence for straightforward PTM site localization. (C) Near complete sequence characterization was achieved with the EAD dissociation.
EAD fragmentation was performed across a range of kinetic energies, taking advantage of the tunable nature of EAD on the ZenoTOF 7600 system. This allowed determination of the optimal fragmentation parameters to preserve labile modifications on PTM modified peptides, while achieving best sensitivity. Kinetic energies were ramped from 0 eV to 11 eV. In this study, it was observed that a KE value of 5 eV generated high intensity, intact PTM site-specific fragment ions with very little background noise. When increasing KE to 8 eV or higher, MS/MS spectra became more complex, and neutral losses from fragment ions containing the PTM modification emerged (Figure 3).
Interestingly, different dissociation patterns were observed for fragment ions across the different kinetic energies during EAD MS/MS fragmentation (Figure 4). The peak area for most fragment ions containing the modified residue and the labile malonyl modification was highest at a KE value of 5 eV. With higher KEs some neutral PTM losses were observed, but a KE value of 5 eV maintained the integrity of the post-translational modification and maximized peak area intensity for site specific fragment ions.
Figure 3. Impact of the kinetic energy (KE) value in EAD on fragmentation pattern. MS/MS spectra of malonylated peptide TVDGPSGKmaLWR (m/z 656.3320) analyzed in EAD mode with a kinetic energy value of (A) 5 eV and (B) 8 eV.
Figure 4. Kinetic energy ramping for EAD MS/MS. Malonylated peptide TVDGPSGKmaLWR was analyzed in EAD mode with kinetic energy (KE) values varying from 0 eV to 11 eV. Chromatographic peaks were extracted for 16 ions. For the intact ions (two first lines), peak area values were normalized on the highest area. For the ions with a neutral loss (third line), area values were normalized to their respective intact ion.
The impact on MS/MS spectral quality and sensitivity when using the Zeno trap was also investigated. The Zeno trap is located in the back half of the collision cell and is used to reduce ion losses between the collision cell and the accelerator. Ions are captured in the Zeno trap and then ejected in order of high m/z to low m/z, such that each ion reaches the center of the TOF accelerator simultaneously. This increases the duty cycle to greater than 90% through this region and greatly increases the sensitivity for MS/MS acquisition.3
EAD MS/MS spectra, collected with and without the Zeno trap activated, illustrate the clear gain in sensitivity provided by this feature (Figure 5). The ratios of extracted fragment ion peak areas, obtained with and without the Zeno trap activated, were determined for PTM site-specific ions at two different amounts loaded (Table 1). The sensitivity gain ranged between 3.5 and 9.6-fold for the various fragment ions, with an average value of 6.4. This highlights the added value of the Zeno trap for investigating very low-abundance ions.
Figure 5. Increased sensitivity when the Zeno trap is activated. MS/MS spectra of TVDGPSGKmaLWR peptide (m/z 656.3320) analyzed in EAD mode (KE = 5 eV) (A) without and (B) with Zeno trap activated. Significant improvements in signal intensity and thus spectral quality was observed while using the Zeno trap in combination with EAD, enabling more confident PTM localization.
Table 1. Gain of MS/MS sensitivity using the Zeno trap. TVDGPSGKmaLWR peptide was analyzed in EAD mode (kinetic energy = 5 eV) with and without using the Zeno trap at various amounts (16 and 80 fmol) on column. Chromatographic peak areas were extracted and sensitivity changes between having the Zeno trap on and off were determined.
To gain first insights into the quantitative performances of EAD MS/MS, initial dilution curves were generated by performing triplicate injections of 4 different peptide concentrations (16, 80, 400 and 2000 fmol on column) (Figure 6).
Good linearity performances were achieved for 9 investigated fragment ions (R2 ≥ 0.99) in these initial assessments.
Figure 6. Linear response of fragment ions during EAD MRMHR. Peptide TVDGPSGKmaLWR was analyzed in EAD mode (KE = 5 eV), in triplicate, loading various amounts (16, 80, 400 and 2000 fmol). 9 fragment ions are displayed. The R2 coefficient for determination of the linear regression is displayed for each ion.
In this work, the utility of electron activated dissociation on the SCIEX ZenoTOF 7600 system was investigated for the characterization and quantification of labile post translational protein modifications.
The Destination for All Your Support Needs